Microlife® WatchBP® O3 (Out-Of-Office): a highly affordable, clinically-validated ambulatory blood pressure measurement device.
Out-of-office measurements are highly recommended as an adjunct to office measurements by almost all hypertension organizations. The WatchBP O3 removes many obstacles generally associated with out-of-office measurements.
In addition to providing a highly affordable ambulatory blood pressure measurement device (ABPM), the WatchBP O3 also includes an embedded 7-day home measurement mode that strictly follows ESH and AHA measurement guidelines. As such, the WatchBP O3 serves as a total out-of-office blood pressure measurement device, validated and recommended for clinical, home and 24-h blood pressure monitoring (http://www.dableducational.org).
The WatchBP O3 has sophisticated software free of charge.
With 5 year warranty replacement*
- 24-Hour ABPM Mode - Fully programmable 24-hour blood pressure measurement.
- 7-Day SBPM Mode - for collecting reliable self-measurement data.
- Lightweight and Compact - For improved patient comfort.
- PC Connectivity - Transmit BP measurement data to any PC via USB connectivity.
Operating temp.: • 10 - 40 °C / 50 - 104 °F
Storage temp. • -20 - +50 °C / -4 - +122 °F • 15 - 90 % relative maximum humidity
Weight: • 260g (including batteries)
Dimensions: • 115 x 80 x 35 mm
Measuring procedure: • Oscillometric, corresponding to Korotkoff
Measurement range: • 30 - 280 mmHg – blood pressure • 40 - 200 beats per minute – pulse
Cuff pressure display: • Range: 0 - 299 mmHg • Resolution: 1 mmHg • Static accuracy: pressure within ±3 mmHg • Pulse accuracy: ±5 % of the readout value
Voltage source: 4X1.5 V Batteries: size AAA
Reference to Standards:
• EU Directives 93/42/EEC
• NIBP requirements: EN 1060-1 / -3/ -4, EN 60601-2-30, ANSI /AAMI SP10
• 550MHz CPU. 256MB Memory
• 1024x768 pixel resolution
• 256 color, CD-ROM drive
• 1 free USB port
• 40MB free hard disk space
• Microsoft Windows
*Terms and Conditions apply - call for details.
The vision on this model is brill! Very sturdy and ergonimic, would purchase again!Posted by Dr Dua | 02 April 2013 More
Great price. Made an offer on the item and got 10% off this item. Glad I got this. Most reliable Opthalmoscope available. Speedy service!Posted by Mr Coltyer | 09 May 2013 More